The KE5FX GPIB Toolkit is a collection of free Windows utilities that will help you make and record research-quality measurements with GPIB-based electronic test equipment.
This is version 1.906 of the Toolkit, released April 16, 2013. The latest release is always available for downloading here. For troubleshooting help and additional application notes, check the FAQ.
The GPIB Toolkit is provided with full C++ source code for public- and private-sector, educational and Amateur Radio / hobbyist use. Comments and feedback are always welcome.
John Miles, KE5FX
7470.EXE is a Win32-based emulator for the HP 7470A plotter. See its User Guide for more information.
PN.EXE is an automated phase-noise measurement utility for HP, Tektronix, and Advantest spectrum analyzers.
See its User Guide for more information.
SSM.EXE is a spectrum surveillance monitor application for HP, Tektronix, and Advantest spectrum analyzers, as well as Signal Hound USB spectrum analyzers. See
its User Guide for more information.
PROLOGIX.EXE allows you to select and configure a Prologix GPIB-USB or GPIB-ETHERNET adapter for use with the GPIB Toolkit applications. It can also be used to select a National Instruments adapter.
All of the GPIB Toolkit applications support both Prologix and National Instruments GPIB adapters. By default, the Toolkit expects to find a National Instruments device at GPIB0. When using a Prologix adapter, it's necessary to edit the CONNECT.INI file to tell the GPIB Toolkit where to find the Prologix adapter's virtual serial port or network address.
This task is easy to accomplish with PROLOGIX.EXE. Simply run the program, select the Prologix or National Instruments adapter in the device list, and press the Update CONNECT.INI button. (Be sure to exit from PROLOGIX.EXE before attempting to run any other GPIB Toolkit application.)
Beginning with the V1.50 release of the GPIB Toolkit, it is no longer necessary to select Device or Controller mode in PROLOGIX.EXE unless you wish to use the optional Terminal or Advanced Settings controls. Each Toolkit application will configure the Prologix adapter's mode and address settings as needed.
Usage examples: prologix Run with no command-line options for normal enumeration of all Prologix GPIB-USB, GPIB-ETHERNET, and National Instruments adapters prologix -noni Don't attempt to enumerate National Instruments adapters (useful if error messages are reported after a partial NI488.2 driver uninstallation) prologix -nonet Don't issue UDP broadcasts to detect Prologix GPIB-ETHERNET adapters
8753.EXE allows you to save and load control/calibration states for the HP 8753A/B/C/D/E/ES vector network analyzers. It can eliminate the need for a disk drive in many cases.
Release 1.00 of 5-Feb-13
SATRACE.EXE can be thought of as a console-oriented version of SSM. It requests one or more traces from any spectrum analyzer supported by SSM, and writes them to stdout as lists of comma-separated frequency/amplitude pairs.
SATRACE's command-line parameters are very similar to those supported by SSM.
Usage: satrace <address> [<options>...] Examples using GPIB address 18: satrace 18 Auto-identify analyzer at GPIB address 18 and acquire a single trace from it satrace -sa44 Special option required for use with USB-SA44/B or USB-SA124A Signal Hound (No GPIB address is needed) satrace 18 -856xa Special option required for use with HP8566A-HP8568A satrace 18 -8569b Special option required for use with HP8569B/8570A satrace 18 -358xa Special option required for use with HP3588A/3589A satrace 18 -3585 Special option required for use with HP3585A/B satrace 18 -advantest Special option required for use with supported Advantest R3200/R3400-series analyzers satrace 18 -r3261 Special option required for use with R3261/R3361 satrace 18 -scpi Special option required for use with Agilent E4400- series and other SCPI-compatible analyzers satrace 18 -f Favor speed over resolution, CRT updates, or other factors (if supported) satrace 18 -t Disable GPIB timeout checking during long sweeps satrace 18 -ao:-7 Add -7 dBm to all reported amplitude values satrace 18 -fo:150000000 Add 150 MHz to all reported frequency values satrace 18 -reps:15 Acquire 15 successive traces (0 = run until keypress) satrace 18 -header Display time/datestamp and available analyzer control settings satrace 18 -lf Separate frequency/amplitude pairs with linefeeds rather than commas satrace 18 -spline:800 Resample trace using cubic spline reconstruction to generate (e.g.) 800 points, regardless of the analyzer's trace array width satrace 18 -point:128 Resample trace using point-sampled values satrace 18 -min:128 Resample trace using minimum bucket values when more source points than requested are available (otherwise use spline) satrace 18 -max:128 Resample trace using maximum bucket values when more source points than requested are available (otherwise use spline) satrace 18 -avg:128 Resample trace using averaged bucket values when more source points than requested are available (otherwise use spline) satrace 18 -connect:"xxx" Specify GPIB command string to be issued when initially connecting to instrument satrace 18 -disconnect:"xxx" Specify GPIB command string to be issued when disconnecting prior to program termination satrace 18 -856xa -f Example using multiple optionsLike SSM, SATRACE does not normally attempt to control or configure the spectrum analyzer. Traces are acquired based on the current front-panel settings such as center frequency, reference level, resolution bandwidth, and so forth. However, SATRACE does provide limited control options for certain analyzer models. These options are shown below.
Additional control options supported by Signal Hound only: -RL:-40 Specify reference level in dBm (default = -30) -CF:90.3E6 Specify center frequency in Hz (default = 900 MHz) -span:10E6 Specify span in Hz (default = 1 MHz) -start:88E6 Specify sweep start frequency (default = 899 MHz) -stop:108E6 Specify sweep stop frequency (default = 901 MHz) -bins:64 Specify FFT kernel size (power of 2 from 16 to 256, default = 128) -sens:2 Specify sensitivity factor (default = 2, range 0-2) -RFATT:10 Specify RF attenuation in dB (0-15 dB in 5-dB steps, default = 0)In most cases, SATRACE's output will be "piped" to another file or application. For example, the command
c:\program files\ke5fx\gpib>satrace 18 -reps:100 -point:500 >temp.txtwill read 100 successive traces from the analyzer at GPIB address 18, point-sample them to generate 500 (Hz,dBm) value pairs for each trace, and write the resulting comma-separated value lists to the file temp.txt.
For more details on trace resampling, see the FAQ ("How do the trace-resampling options in SATRACE and SSM work?").
Release 1.01 of 25-Jul-08
SGENTEST.EXE tests HP signal generators (8656A/B, 8757A-D, 8642B/M, 8662A/8663A and others) by programming randomly-chosen frequency and amplitude settings. Output signals are then verified by an HP 8566A/B, 8568A/B, or 8560A/E-series spectrum analyzer.
Usage: sgentest <analyzer address> <generator address> [<options> ...] Available options (with default or example values): -fmax:1.2E9 .... Maximum test frequency = 1200 MHz -fmin:10E3 .... Minimum test frequency = 10 kHz -amin:-90 .... Minimum test power = -90 dBm -amax:10 .... Maximum test power = 10 dBm -loss:1.0 .... Interconnect loss in dB at fmax -ftol:200 .... Frequency tolerance = +/- 200 Hz -atol:2 .... Amplitude tolerance = +/- 2 dBm -wait:0.25 .... Wait 250 milliseconds after programming generator -8662 .... Check for HP 8662A/8663A hardware errors -8560 .... Compatibility mode for certain 8560A/B/E portable analyzers -log:filename .... Log output to filename -trials:1000 .... Stop after 1000 trials -verbose .... Display/log all trials, not just failures -srand:500 .... Seed random # generator with integer -stime .... Seed random # generator with time() function Notes: - Minimum test power must be sufficient to yield good amplitude measurements - Numeric arguments are reals, and may use scientific notation - 8662A/8663A hardware error checks are disabled by default - Some 8560A/B/E-series portables may require -8560 switch for compatibility (which inhibits use of KSX and KSb commands for amplitude correction and positive-peak detection) - Test will run indefinitely if -trials is 0 (or not specified) - Default -loss value is based on 15 feet of LMR-400 and two N connectors at fmax=1.2E9 Example to test HP 8663A at address 19 with analyzer at address 18, using 15 feet of LMR-400 cable (1.0 dB loss) with N connectors (0.15 dB each): sgentest 18 19 -fmax:2559E6 -amax:19 -8662 -loss:1.3\n");
By default, the following utilities are installed with the rest of the GPIB Toolkit executables in the C:\Program Files\KE5FX\GPIB directory. In most cases you can obtain usage notes by running each program with no command-line arguments.
PARSE.EXE is a simple command-line utility that allows you to inspect HP-GL files by dumping their instructions to stdout.
QUERY.EXE is a standalone GPIB query utility intended for diagnostic purposes and batch/shortcut files.
TALK.EXE is a standalone GPIB command-transmission utility intended for diagnostic purposes and batch/shortcut files.
LISTEN.EXE is a standalone GPIB data-reception utility for ASCII traffic, intended for diagnostic purposes and batch/shortcut files.
BINQUERY.EXE is a standalone GPIB query utility intended for diagnostic purposes and batch/shortcut files. It illustrates how to retrieve arbitrarily-large binary data blocks from GPIB instrumentation.
GPS.EXE is an example program for the GNSS.DLL library. It displays incoming latitude, longitude, and altitude fixes. GNSS.DLL provides a C API for NMEA-compatible GPS receivers such as the u-blox LEA-6 series.
5345A.EXE and 5370.EXE are two example programs that show how to communicate via GPIB with the HP 5345A and HP 5370A/B frequency counters, respectively. They are meant to be modified and recompiled as needed.
DTS2070.EXE demonstrates capture of timing data from a Wavecrest DTS2070 timing analyzer. Like the HP counter utilities it is meant to be modified as needed.
DSO6000.EXE retrieves voltage data from a trace captured on an Agilent DSO/MSO6000A-series oscilloscope. If an output filename is supplied on the command line, the data will be written as a series of double-precision binary values; otherwise, the voltage values will be printed to stdout. DSO6000.EXE assumes that a SINGLE sweep has run to completion, and that the oscilloscope is in STOP mode (STOP/RUN button illuminated red.) It was tested on an MSO6054A, and may or may not be compatible with other Agilent models such as the DSO/MSO5000A series.
CAL_TEK490.EXE contains various calibration aids from the Tektronix 492P service manual. Run the program for a menu of available calibration routines.
ECHOCLIENT.EXE and ECHOSERVER.EXE are intended for troubleshooting and benchmarking TCP/IP connections. These utilities communicate over port 4242. (Note that benchmark results obtained from these programs are normally limited by the GPIB Toolkit's network layer, rather than by the connection itself.)
PRINTHPG.EXE sends a plotter data file (.PLT, .HPG, etc.) to the default DOS printer (PRN device). This printer should be an HP LaserJet or compatible model that supports legacy HP-GL commands. Other printers can be used by 7470.EXE, but the bitmaps printed by 7470.EXE will lack the resolution and fidelity obtainable by sending the plotter file directly to a genuine HP-GL-compatible printer.
If you have a LaserJet and want the best-quality black-and-white output possible, PRINTHPG.EXE is the way to go. Note that even LaserJets don't support some HP-GL features, like UC (User Character) opcodes! If your instrument relies on these, you'll need to use 7470.EXE to render its plots and print them as fixed-resolution bitmaps.
TCPLIST.EXE transmits a string to a specified TCP/IP address and listens for a reply. Command-line options specify the timeout interval, EOS character, and the number of ASCII text lines expected in response. TCPLIST.EXE is handy for accessing instruments that support Telnet connections or other simple ASCII-based protocols.
By default, the following batch files are installed with the rest of the GPIB Toolkit executables in the C:\Program Files\KE5FX\GPIB directory. They may be inspected and modified with any word processor or text editor capable of working with plaintext ASCII files.
WIFI_HP8566.BAT uses TALK.EXE to program an HP 8566B spectrum analyzer at GPIB address 18 for 2350-2550 MHz sweeps at 5 dB/division, selecting a -35 dBm reference level and 0 dB of RF attenuation. It then launches SSM.EXE with max-accumulation and WiFi band boundaries enabled.
WIFI_TEK490.BAT uses TALK.EXE to program a Tektronix 492P/494P/494AP/497P analyzer at GPIB address 3 in a similar fashion. The analyzer's CRT readout is turned off to prevent burn-in.
WIFI_E4406A.BAT programs an Agilent E4406A Vector Signal Analyzer at GPIB address 18 to display the middle 10 MHz of WiFi channel 1 (2412 MHz). It then launches SSM.EXE with max-accumulation and WiFi band boundaries enabled.
GPS_SA44.BAT programs a Signal Hound USB-SA44/B or USB-SA124A spectrum analyzer to display the middle 6 MHz of the GPS L1 band at 1575.42 MHz. It then launches SSM.EXE with max-accumulation and GPS band boundaries enabled.
R3267.BAT uses BINQUERY.EXE to request (.BMP) images from an Advantest R3264, R3267, or R3273 spectrum analyzer at GPIB address 7.
IMD_8566B_8568B.BAT uses TALK.EXE to transmit the downloadable program (DLP) T_HIRDIMOD to the nonvolatile user RAM on an HP 8566B or 8568B spectrum analyzer. This program performs basic third-order intermodulation distortion measurements on signals between 10 MHz and 500 MHz, and also calculates IP3. See HP product note 8566B/8568B-1 for more information.
ESM.BAT uses TALK.EXE to control a Rohde & Schwarz ESM-500A receiver. Run the batch file without any arguments to see a brief usage guide.
TSCCMD.BAT uses TCPLIST.EXE to control a Symmetricom TSC 5115A/5120A/5125A phase noise test set. Run the batch file without any arguments to see a brief usage guide.
TCP2PNP.BAT uses TCPLIST.EXE to save the phase noise spectrum from a Symmetricom TSC 5115A/5120A/5125A as a .PNP file compatible with the PN.EXE phase noise utility.
HP 8753 S2P Downloader, contributed by Melvin Clarkson of Aspire Design Ltd, fetches Touchstone .S2P S-parameter files from an HP 8753 A/B/C network analyzer equipped with a Prologix USB adapter. VB 2010 source is included. These .S2P files may be imported into circuit simulator programs including the free RFSim99 package.
To use the program, unzip the file and run bin\Debug\SerialPortCommunication.exe. Select the Prologix adapter's COM port, then press the "Open Port" and "S paras" buttons.
The C:\Program Files\KE5FX\GPIB directory also contains three subdirectories, Composite noise baseline plots, Sample HPGL files, and Time interval plots. These subdirectories contain a variety of .PNP and .PLT files that can be loaded and displayed by PN and 7470, respectively.
Released into the public domain 16-Dec-05 by John Miles, KE5FX. Please contact firstname.lastname@example.org with feedback and enhancements!
Some DSPLIB components used by the GPIB Toolkit are subject to open-source license agreements; see the current DSPLIB distribution for details.
Source code compilable with the free Microsoft Visual Studio Express package
No warranties, expressed or implied, are offered with these programs.
Use at your own risk!