The graphs below compare the phase noise and stability characteristics of three BVA DOCXOs. These include two 8607/08 models (designated TVB and JRA) and one 8607 model that meets option-67 specifications (designated EWK).

See AN 53100A-2, Dual Reference Noise and Stability Measurements with the PhaseStation 53100A (2 MB .PDF) for notes on the test methodology, as well as the PhaseStation 53100A User's Manual (4 MB .PDF).

.TIM files for the initial multichannel runs are available here (24 MB).

The test mask in the plot above corresponds to the data on page 4 of the OSA factory test report for the

Beyond t=60 seconds the three-cornered hat solution becomes highly questionable for **JRA**. It is probably reasonably accurate for
the **EWK** unit, but it is completely invalid for **TVB**.

Several repetitions of this 8-hour test with the same OCXOs yielded very similar traces, but I'm still not especially satisfied with the quality of the results in this initial trial. More testing is needed (see further comments below).

The phase noise results are a good match for those in the test report, as well as the Rubiola citation mentioned in AN 53100A-2.

This plot is a simultaneous three-channel measurement of the 8607 BVAs against a single reference, an HP 5061A cesium-beam standard. The

*Edit:* At least, it was unclear before going back and examining the data for the October 2, 2019 entry below. The traces above were captured on September 29, at which point the
EWK and TVB units had been powered on for approximately five days. Unsurprisingly, this isn't enough time for the oscillators to reach their full performance potential.

The frequency-difference plot corresponding to the measurement above appears here.

This may help explain TVB's degenerate variance in the earlier three-cornered hat measurement. The algorithm makes a reasonable guess at the variance of the EWK unit, but it does a poor job at characterizing the variance of the JRA unit since it's lower than either EWK or TVB, with the difference dominated by linear drift. Consequently the variance of the TVB unit can't be measured reliably beyond t=60s.

We can conclude that the three-cornered hat algorithm is probably doing a good job at taus below 60 seconds, but only the highest-variance oscillator is being characterized even remotely accurately at longer taus.
It's interesting to note that the measurement with the lower-quality 8600B OCXO in AN 53100A-2 actually yielded a better three-cornered hat solution. This is likely because the TVB 8607 oscillator had been
running for a much longer period of time when the AN 53100A-2 data was collected.

Startup behavior of the

This series of 8-hour runs was taken across several days, showing the TVB and JRA oscillators measured directly against each other.

Notes:

- The phase records have been scaled by 0.707 based on the assumption that these two oscillators are fairly close in performance.
- Below t=30s, the
**JRA**and**TVB**traces in the previous three-cornered hat plot are a good match for these independent measurements. - Beyond t=30s, the three-cornered
**JRA**trace is probably more accurate than I'd initially assumed it was. - In principle, the earlier three-cornered
**TVB**trace should have closely resembled the one above from**September 29**, which is the date when the three-cornered hat measurement was performed. - It looks like the stability of the TVB unit beyond t=100s has improved a great deal since it was last powered up for the warmup tests conducted on September 24. The improvement over the September 27-October 2 timeframe is not only monotonic, but seems to be accelerating.
- Given its reduction in drift over time, it seems likely that a repeat of the three-cornered hat plot will show better results for the TVB oscillator than what we observed earlier. Watch this space...
- The PhaseStation with serial #58526.447444 is a modified preproduction unit with an extra 6 dB of attenuation in its DUT input splitter for channels 3 and 4. Consequently, the short-term differences near t=0.1s should be disregarded.

The frequency-difference traces corresponding to the ADEV measurements above reflect the stability improvement as well.

Data for these additional runs can be downloaded here (15 MB).